Operating Instructions for TE-302 Digital Timer
To set up the digital timer:
Start with the Sampler Switch (Timed – Off – On) Switch #1, in the
If you need to test or adjust the blower motor turn the Sampler
switch to On. When done with adjusting, turn it back to Off.
Place the rotary switches in the desired positions.
If today is Friday and you want the first sample time on Sunday, turn
the “Sample After Days” switch to position 2.
If you want to run the sampler every Sunday after that, turn the
“Sample Every Days” switch to position 7, (for six day sampling use
Turn “Sample for Hours” to desired number of running hours.
Next put the Display switch, Switch #4, in the Start Time position.
Then using the Set switch, Switch #3, enter the start time, hours and
Next put the Display switch, Switch #4, in the Time of Day position.
Then using the Set switch, Switch #3, enter the current time, hours
Now press and release the Reset switch, Switch #2, toward Timer. A
small triangle on the display will start blinking. This indicates the
timer is running.
If you need to reset the Hour Meter to zero.
Press and release the reset switch, Switch #2, twice, toward Hour
Last thing to do is place the Sampler switch, Switch #1, (Timed – Off
– On) in the Timed position