6. Locate hood spacer between hood and acceleration nozzle plate and
loosely fasten with 10-32 x ½” thumb bolt, making sure plastic washer
is in place. Do this loosely for all eight hood spacers, before
7. Open TE-6001 SSI by disengaging hooks and lifting the middle section
into the open position. Remove cardboard and rubber bands that are
covering filter holder assembly opening.
8. Screw Filter Holder on to VFC Device, be sure gasket is in place.
9. Lower filter holder assembly down through opening, making sure 8” x
10” gasket is under filter holder.
10.Put TE-3000 Filter Paper Cartridge on top of filter holder and align
the brass bolt assembly with the cartridge. Tighten for airtight
IMPORTANT: Remove cover on top of TE-3000 Filter Paper Cartridge
before turning on sampler. The cover is only used to protect
sample from contamination during transport.
11.Connect clear piece of tubing from inside of shelter on to brass
pressure tap located on the filter holder side.
12.Close Inlet, making sure of an airtight seal.
13.Before operating, make sure TE-6001-24 Shim Plate has been wiped
clean and then treated with Dow Corning Silicone spray 316, evenly.
(See Sampler Operation)
ELECTRICAL HOOK-UP TE-6070
M F M F M F F M