TE-WILBUR OPERATIONS MANUAL REV 002.15 06/11/2015
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 146 of 183
Open front cover, if green LED is not ON check
voltage on black connector, if voltage is
present, check operation of the pushbutton and
ensure voltage to the purple connector when
button is pressed, replace green LED
pushbutton if not functioning properly.
If voltage is not present on the black connector
and mains power is plugged in, then check the
1A circuit breaker for proper operation.
No DC power
Check DC operation – ensure AC mains power
is present on L and N on the top of the power
supply and DC(+) and DC(-) has 27VDC
present. If DC power is present and the unit
will not power on, check and replace the 3A
If there is AC voltage present on the power
supply and no DC voltage from the power
supply, disconnect the DC(+) terminal and
measure again. If still no voltage replace power
If after disconnecting the DC(+) wire and there
is 27VDC on the DC(-) and DC(+) terminals,
there is a short somewhere between the DC(+)
and DC(-) terminals.
Sampler states ‘mains
power off’ but unit is
plugged into mains
Power fail relay
Ensure mains power is on and by following
steps above. There is a green LED on the front
of the relay that indicates the unit is powered.
Check to ensure there is AC voltage on
terminals A1 and A2 on the power fail relay.
Check continuity on terminals 11 and 12.
When power is off, these terminals should be
closed and when power is on the terminals
should be open.
If power fail relay is operating properly, check
the input to the PLC. The power fail relay is
connected to input 3. When mains power is off
the LED on the PLC for input 3 should be ON.