TE-WILBUR OPERATIONS MANUAL REV 002.15 06/11/2015
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 64 of 183
3. Place a filter cassette with a screen and filter into the filter holder and
close the filter holder. NOTE: this filter cannot be used for subsequent
sampling per USEPA Quality Assurance Guidance Document 2.12.
4. Proceed to the setpoint screen where the user can change the setpoint
of the flow calibration. 16.67 Lpm is the default setpoint and should
only be changed under abnormal operations.
Flow Calibration Setpoint Screen
5. Press START and the Calibration Point 1 screen will be displayed and
the flow system will start and will achieve the first setpoint. The
setpoint and flow is shown on the left.