TE-WILBUR OPERATIONS MANUAL REV 002.15 06/11/2015
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 65 of 183
Flow Calibration Enter Reading Point 1 Screen
6. After the flow stabilizes, take a reading from the calibrator and press
the blue box to enter the calibrator’s reading.
7. Press the green ACCEPT box to accept that reading and continue to
setpoint number two.
8. Perform setpoints two, three and four. After the fourth setpoint the
final flow calibration screen will appear.
NOTE: If the QUIT RETURN TO MAIN MENU Button is pressed on any
of the screens, the last successful calibration slope and intercept will
be loaded into the flow calibration settings and the calibration will be