TE-WILBUR OPERATIONS MANUAL REV 002.15 06/11/2015
Tisch Environmental 1-TSP-AND-PM10 firstname.lastname@example.org www.tisch-env.com Page 75 of 183
8.3 Battery Disconnect
In order to power down the sampler the battery must be disconnected before
power is removed. To disconnect the battery, press the button that is labeled:
PRESS FOR BATTERY DISCONNECT.
NOTE: If mains power is removed and the sampler is running on battery,
pressing this button will power down the sampler. Mains power
must be applied to the sampler to power on again.
8.4 Manual Control
In the manual control screen the system can be operated in automatic flow
control mode by entering a flow setpoint or can be operated by adjusting the
percent of speed of the pump with no flow control.
The manual control screen is found here:
To run the pump with automatic flow control, perform the following:
1. Enter the setpoint by pressing the setpoint box and entering in the flow
setpoint for automatic flow control.
2. Press the START button underneath the flow setpoint.
3. The system will turn on and the pump will operate at the desired
4. To stop the system press the STOP button below the setpoint box.
Manual Control Screen showing pump in flow control mode