TE-WILBUR OPERATIONS MANUAL REV 002.15 06/11/2015
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 76 of 183
To operate the pump manually with a percent of speed, perform the following:
1. Press the MANUAL START BUTTON.
2. Press the right arrow button to increase the speed of the pump.
3. Press the left arrow button to decrease the speed of the pump.
4. Press the MANUAL STOP button to stop the pump.
Manual control screen showing pump in manual % speed control
NOTE: If the system is running in manual mode and a sample starts, manual
mode will be shut down and the sample will start as scheduled.
NOTE: When running in percent of pump speed manual mode, the only alarm
that will shut down the pump is the high pressure alarm which is set
to 200” H2O of pressure.
NOTE: When running in flow control manual mode, all the alarms that will
shut down a sample will also shut down the automatic flow control in
manual mode with the exception of the flow variance alarm and the
ten consecutive power fails alarm. See section 1.4 – Sample Alarms