TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 firstname.lastname@example.org www.tisch-env.com Page 112 of 185
Powering off the system
In order to power off the system, the batteries must be disconnected. To
disconnect the batteries perform the following keystrokes:
Pressing this button will disconnect the batteries from the system and allow the
system to be powered down completely by removing mains power.
NOTE: If mains power is removed and the sampler is running on battery,
pressing the disconnect battery button will power down the sampler.
Mains power must be applied to the sampler to power on again.
The mains electrical connection is a standard IEC-320 plug configuration and
is found in the weatherproof box on the side of the sampler.
A 1Amp resettable circuit breaker is located just below the AC inlet plug. If
this circuit breaker would trip, it can be reset by pushing the button on the front.
WARNING – When powering down the system to perform maintenance, it is
important to remove mains power from the sampler by unplugging the cord
from the power source. The terminals for the plug and circuit breaker will be
energized and could be accidentally touched inside the unit, resulting in a shock