TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 157 of 185
If the fan does come on when placing the jumper
on terminal 3, check the wiring to the PLC card
(outputs right side). When the fan is supposed
to be ON the LED for output 4 should be on.
Replace PLC card.
14.11 Lighting Strip
The LED lighting strip is TE-Wilbur part number TE-W-59-R for the right side
of the enclosure and TE-W-59-L for the left side of the enclosure.
LED Lights do not
come on when button
Take a jumper wire and place one end on
24VDC and the other end on terminal 1. The
lighting strips should come on. Check for
27VDC on the connector to each lighting strip.
If the LED lighting strip(s) do not come on
check the wiring. Replace lighting strip(s).
If the LED lighting strips do come on when
placing the jumper on terminal 1, check the
wiring to the PLC card (outputs right side).
When the lighting strip(s) are turned on, the
LED for output 3 should be on. Replace PLC
14.12 Solar / Alternative Energy
For help with troubleshooting issues with the solar / alternative energy options
see section 12.0 Solar and Alternative Energy.
For help with troubleshooting communication issues with the TE-Wilbur
sampler, see section 15.0 Communications.