TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 firstname.lastname@example.org www.tisch-env.com Page 27 of 185
3.0 Quick Start Guide
The TE-Wilbur system comes from the factory with the following functions
o Leak Check Passed
o Flow System Calibrated
o Ambient Temperature Calibrated
o Filter Temperature Calibrated
o Barometric Pressure Calibrated
o Pressure sensor zeroed
o 48-hour burn-in test running at 16.67 Lpm
o Time and Date set for Eastern Time (UTC -5.00, Eastern Time)
Perform the following to place unit in service:
Unpack Unit from Shipping Box.
Determine sampling location
Place stand on level surface and secure stand with sand-bags or screw onto a fixed
surface. Place the sampler onto the stand and secure with (8) ¾” ¼-20 bolts (included)
and mounting brackets.
Secure the ambient temperature radiation shield / probe onto left side of unit with (2)
¾” ¼-20 bolts (included) and screw the probe connection into the socket.
Place the Cyclone on top of the filter holder, place the downtube through the downtube
adapter and push onto the top of the cyclone (if configuring for PM10 do not place the
cyclone onto the filter holder) Leave the PM10 head off for now in order to perform a
Plug the unit into an AC power source and press the green button to power on unit.
Unit should power on – if problems see Troubleshooting – Section 13.0.
At the splash screen, choose your language.
If you are not in EST time zone, change the time and date if necessary by going to:
MAINTENANCESET TIME AND DATE.