TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 28 of 185
Perform an external leak check by placing the L30 leak check adapter on top of the
down-tube and closing the valve. Then go to:
MAIN MENUPERFORM LEAK CHECKS
Perform a flow calibration check of the flow system by going to:
CALIBRATIONFLOW CALIBRATION CHECK
If the flow is within 4% no action is needed. If flow is not accurate, then a flow
calibration must be performed. See section 6.5.
Check the barometric pressure, filter temperature and ambient temperature sensors, the
values can be found by going to:
MAIN MENUOPERATIONAL DATA
If the sensors are not reading correctly, a calibration must be performed see section 6.0
Enter the Site ID and Filter ID by going to:
SAMPLE SETUPENTER SITE AND FILTER ID
Choose the sample you would like to perform and the starting day:
SAMPLE SETUPSET TO SAMPLE 1 in 6 DAYS (for example)
Place the PM10 head onto the top of the downtube.
Place a clean filter into the filter holder and close the filter holder.
Close the enclosure of unit.
After sampling is complete, insert a USB thumb drive into the USB port and press
download run summary data. After data downloads press ‘EJECT USB’ to eject the
USB drive and then remove the USB drive.
Remove dirty filter and record all data as required by your agency.
Place a new filter in the filter holder and close the filter holder.
Update the filter ID and ensure the unit is still setup for the sample type you are running.
The sample control screen will show the next date of the sample..