TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 firstname.lastname@example.org www.tisch-env.com Page 44 of 185
5.0 PM10 Inlet / PM2.5 Cyclone
This section describes the TE-PM10 inlet and the TE-PM2.5C PM2.5 cyclone
5.1 TE-PM10 Inlet
The TE-PM10 inlet is a 10 micron particulate fractionator. Below is a cut-away
view of the TE-PM10.
Ambient air is pulled in through the top of the inlet and particles that have
aerodynamic diameters greater than 10um are impacted on the inside plate of
the unit. The particles that have less than 10um of aerodynamic diameter are
expelled at the bottom, onto the next stage.