TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 firstname.lastname@example.org www.tisch-env.com Page 50 of 185
The 1 in 12 schedule will be shown purple, The 1 in 6 will be shown in green
and the 1 in 3 sample schedule will be shown in orange.
The 1 in 3 sample setup is found here:
SET TO SAMPLE
1 in 3 DAYS
On this screen you have the ability to set a start Month / Day / Year. The system
will not allow you to enter a date in the past.
The unit will start on Midnight of the day you enter. So if you enter 2/23/14
and the date is 2/22/14 and the time is 23:59:59, in one second the unit will turn
on and sample for 24 hours midnight to midnight.
You also see the setpoint of the sample. This defaults to 16.67 Lpm which is
the designed setpoint, so this value should never be changed. See Screen
Maintenance section 8.0 to change the setpoint.
After the sample is setup press the PROCEED TO SAMPLE CONTROL