TE-WILBUR OPERATIONS MANUAL REV 002.22 4/27/2017
Tisch Environmental 1-TSP-AND-PM10 email@example.com www.tisch-env.com Page 51 of 185
Sample control screen
On this screen you have the ability to Cancel the current sample and see all
the pertinent sample runtime information such as:
This is the date that the sampler will start running at
This is the date of the next sample. If the current start
date is 2/23/14 and you are running a 1 in 3 schedule,
the next sample run date will be 2/26/14.
This is the actual runtime of the sample. When the
sample starts this value is reset to zero. The currently
displayed value is the last sample’s runtime or the
current runtime if the sampler is running.
This is the time remaining of the current sample. When
a 24 hour sample starts this value will be 24.
This is the total airflow in cubic meters. At 16.67 Liters
per minute, 1 cubic meter per hour of air will pass
through the sampler. In 24 hours, approximately 24
cubic meters of air will flow through the sampler.
Vol Air Flow
This is the actual volumetric flow in Lpm.