TE-5170DV-BLX & TE-5170BLXZ 26 Operations Manual
Use of Look-Up-Table to Determine Flow Rate
(NOTE: Individual Look-Up Tables will vary.)
1. Suppose the ambient conditions are:
Temperature: Ta = 20 oC
Barometric Pressure: Pa = 749.3 mm Hg (this must be station pressure which is not
corrected to sea level)
2. Assume system is allowed to warm up for stable operation.
divided by 2 for an average reading. This is taken with a differential manometer with one
side of the manometer connected to the stagnation tap on the filter holder (or the Bulkhead
Fitting) and the other side open to the atmosphere. Filter must be in place during this
measurement.
Assume that:
Set-up Reading (clean filter): Pf = 12.6 in H2O
Pick-up Reading (dirty filter): Pf = 16.0 in H2O
Pf = (12.6 + 16.0)/2 = 14.3 in H2O.
4. Convert Pf = to same units as barometric pressure.
Pf = 14.3 in H2O / 13.61 x 25.4 = 26.687729 mm Hg
Pf = 26.69 mm Hg
5. Calculate pressure ratio.
Po/Pa = 1 - (Pf/Pa)
NOTE: Pf and Pa MUST HAVE CONSISTENT UNITS