Sonic Tracker II Transducer Replacement
P-32+ and P-32 Operator’s Manual P/ N: 1001547-01
Figure 173: Replace Transducer, Connectors, and Ty-Wrap
5. “Feed” the wires back up into the Sonic Tracker I I and seat the transducer into place (the two wire tabs
have to sit down into the extra deep area of the recess). Place the new “O”-ring around the transducer and
seat firmly between the transducer and the Sonic Tracker II transducer recess (Figure 174 on page 96).
Figure 174: Replace Wires, Transducer, and “O”-ring
6. Place the black retaining ring over the transducer assembly with the beveled edges out and the flat surface
against the Sonic Tracker I I base. Line up the holes for the mounting screws. The mounting screw holes
are set to an irregular pattern, ensuring the retaining ring only lines up with the mounting holes one way
(Figure 175 on page 97).
7. Start the mounting screws into the holes and tighten each until firm (Figure 175 on page 97). DO NOT
OVER TIGHTEN. Use a cross (X) pattern when tightening the screws.
PLACE NEW TY-WRAP
DO NOT use Loctite® on the mounting screws. Loctite® attacks and degrades the plastic retaining
Over tightening the screws distorts the transducer metallic material and may crack the retaining ring.
The O-ring provides tightness while preventing vibration from loosening the mounting screws.