GENERAL INFORMATION ON USAGE AND MAINTENANCE
The product conforms to the EMC standard (IEC 60601-1-2 Ed3.0:2007)
a)MEDICAL ELECTRICAL EQUIPMENT needs special precautions regarding EMC and needs to be
installed and put into service according to the EMC information provided in the ACCOMPANYING
b)Portable and mobile RF communications equipment can affect MEDICAL ELECTRICAL EQUIP-
c)The use of ACCESSORIES, transducers and cables other than those specified, with the exception
of transducers and cables sold by the manufacturer of the EQUIPMENT or SYSTEM as replace-
ment parts for internal components, may result in increased EMISSIONS or decreased IMMUNITY
of the EQUIPMENT or SYSTEM.
d)The EQUIPMENT or SYSTEM should not be used adjacent to or stacked with other equipment. If
adjacent or stacked use is necessary, the EQUIPMENT or SYSTEM should be observed to verify
normal operation in the configuration in which it will be used.
e)The use of the ACCESSORY, transducer or cable with EQUIPMENT and SYSTEMS other than
those specified may result in increased EMISSION or decreased IMMUNITY of the EQUIPMENT or
Guidance and manufacturer's declaration - electromagnetic emissions
The TRK-2P is intended for use in the electromagnetic environment specified below.
The customer or the user of the TRK-2P should assure that it is used in such an environment.
Emissions test Compliance Electromagnetic environment - guidance
CISPR 11 Group 1
The TRK-2P uses RF energy only for its internal function.
Therefore, its RF emissions are very low and are not likely
to cause any interference in nearby electronic equipment.
CISPR 11 Class B
The TRK-2P is suitable for use in all establishments other
than domestic and those directly connected to the public
low-voltage power supply network that supplies buildings
used for domestic purposes.