MEASUREMENT IN REF/KRT→TONO/PACHO CONTINUOUS MEASUREMENT MODE
DISPLAYING MEASUREMENT VALUES
With regard to measurement values, for REF, KRT, TONO and PACHO, data of the latest measure-
ment (only for TONO/PACHO, latest 3 measurements) are displayed on the control panel.
Figures only: Measurement was done correctly.
[ ] figures: When the reliability of measurement is low. (only TONO)
ERROR: Measurement was not done correctly.
OVER: When the measurement range is exceeded. (only TONO)
• In TONO average value display, low-reliability numerical data with [ ] are
not added to average value calculation.
However, if all measurement data are numerical data with [ ], average
value calculation is done using these data.
• For explanation of the messages on the control panel screen, refer to
"MESSAGE LIST" on page 99.
• In the data printout, manual measurement values will have M marks
beside them. (only for TONO/PACHO)
• When the "Auto print" is set to the "OFF" in the "Common" of the "Initial",
print out measurement results by tapping the button, as neces-