3Tap button to exit the full-screen display. The button appears indicating that
the image is saved.
4Tap the button to display the saved image.
5Tap either of the (R)/(L) buttons and move the positioning bar.
6Follow steps 5 to 7 of "MEASUREMENT ON THE ACTUAL IMAGE."
7The cornea diameter is displayed.
8Tap the button or button to move the measuring head to the other eye.
In the like manner, measure the other eye.
9 Tap the button and return to the Measurement screen.
NOTE If retaking still image is required, tap the capture button again.
R: 0 .0 0
L: 0 .0 0
Positioning bar control
R:0.00R: 0 .0 0 L:0.00L: 0 .0 0
R:0.00R: 0 .0 0 L:9.50L: 9 .5 0