SETTING FUNCTIONS ON SETUP SCREEN
In the "REF/KRT" the function common to REF/KRT is set up.
* : Depending on the destination, preset values differ.
Sets the pupil distance between right and left eyes.
(Setting is required when R/L move is "Full Auto" or "Auto(RL)".) 68mm
Auto power save
OFF Power save function is not used.
1min Power save status in 1min after last operation.
5min Power save status in 5min after last operation.
10min Power save status in 10min after last operation.
20min Power save status in 20min after last operation.
30min Power save status in 30min after last operation.
60min Power save status in 60min after last operation.
Date/Time Set by number display Sets year, month, day, time (24hrs), minute and second. Installation
R/L notation R/L Right/left eyes is displayed by R/L. R/L
OD/OS Right/left eyes is displayed by OD(R)/OS(L).
Chin rest height
Default chinrest height. LowCenter
Control panel brightness
LEVEL 1 (dark)
The brightness of control panel is set up. LEVEL 4
LEVEL 4 (bright)
Packing mode Execute Starts packing mode.
Shaded characters ON Font style of measurement values is shaded. ON
OFF Font style of measurement values is not shaded.
XZ MOTOR direction
Adjust The moving direction is changed according
to control panel position. Adjust
Fixed The moving direction is not changed according
to control panel position.
Descriptions Options Details Initial value
Cont. Cycle 1-10
Set by number display The number of continuous measurements. 3
Add Measure 0-99
Set by number display. When the measurement is error,
set the number of times of remeasurement. 1
Continuous fog Every time Continuous fog is applied every time. Once
Once Continuous fog is applied only once before the 1st measurement.
Init. measure mode
REF Default measurement mode on is REF.
REF/KRTREF/KRT Default measurement mode is R/K.
KRT Default measurement mode is KRT.
Manual The switching of right and left eyes is performed at "Manual".
Full AutoFull Auto The switching of right and left eyes is performed at "Full Auto".
Auto (RL) The switching of right and left eyes is performed at "Auto(RL)".
Touch Measure OFF Touch measurement is not performed in Manual mode. ON
ON Touch measurement is performed in Manual mode.
Sph/Cyl step 0.12 Sph/Cyl is displayed by 0.12D step. 0.25
0.25 Sph/Cyl is displayed by 0.25D step.
Axis step 1° Axial angle is displayed by 1° step. 1°
5° Axial angle is displayed by 5° step.
0.00 VD value is set to 0mm (contact lens).
13.75*12.00 VD value is set to 12.00mm (eyeglass lens).
13.75 VD value is set to 13.75mm (eyeglass lens).