SETTING FUNCTIONS ON SETUP SCREEN
In the "TONO/PACHO" the function common to TONO/PACHO is set up.
The typical additional power for the age can be selected. No
D or mm(KRT) D D (diopter) of corneal refractive power. mm
mm mm of corneal curvature.
HV or R1R2
HV Corneal curvature radius measurement result on screen
is displayed by HV. R1R2
R1R2 Corneal curvature radius measurement result on screen
is displayed by R1R2 (flat/steep meridian).
Display KRT unit OFF KRT unit is not shown. ON
ON KRT unit is shown.
-Cylinder sign is "
+Cylinder sign is "+".
MIX Cylinder sign is "+" and "
Picture printer Normal Printer Picture of refractive condition is not printed. Normal Printer
Graphic Printer Picture of refractive condition is printed.
REF average OFF REF average is not displayed. OFF
ON REF average is displayed.
Model Eye Measure Mode Execute Model eye measurement mode will start.
Descriptions Options Details Initial value
Cont. cycle(TONO) 2-10
Set by number display The number of continuous measurements (TONO). 3
Cont. cycle(PACHO) 2-10
Set by number display The number of continuous measurements (PACHO). 3
Add Measure 1-99
Set by number display The number of additional measurements. 1
Low Credibility Measure No Including The value of low reliability is excluded form count cycle. Including
Including The value of low reliability is included to count cycle.
Init. Measure mode TONO Default measurement mode on is TONO. T/P
T/P Default measurement mode is T/P.
Manual Measurement head moves right and left manually.
Full AutoFull Auto Measurement head moves right and left automatically.
Auto(RL) The switching of right and left eyes is performed at "Auto(RL)".
Touch Measure OFF Touch measurement is not performed in Manual mode. ON
ON Touch measurement is performed in Manual mode.
Show Tono value OFF Tono value is not displayed. ON
ON Tono value is displayed.
Show Pacho value OFF Pacho value is not displayed. ON
ON Pacho value is displayed.
Show Tono average OFF Tono average value is not displayed. ON
ON Tono average value is displayed.
Show Adj value OFF Adjusted value is not displayed. ON
ON Adjusted value is displayed.
Tono display Unit
mmHg Display in mmHg
digit Display in digit
hPa Display in hPa
Torr Display in Torr
Pacho display Unit mm Display in mm mm
µm Display in µm
Press average Mode Integer Display in Integer Integer
Real Display in Real
IOP Adjustment OFF IOP adjustment type is OFF. OFF
ON IOP adjustment type is ON.
Center CCT Base 0-999
Set by number display Sets the central cornea thickness base value.
(Used when IOP adjustment is "ON.") 545
Adjustment Coefficient 0-999
Set by number display Sets the adjustment coefficient.
(Used when IOP adjustment is "ON.") 500