This product conforms to the EMC Standard (IEC 60601-1-2:2001).
a) MEDICAL ELECTRICAL EQUIPMENT needs special precautions regarding EMC and
needs to be installed and put into service according to the EMC information provided in the
b) Portable and mobile RF communications equipment can affect MEDICAL ELECTRICAL
c) The use of ACCESSORIES, transducers and cables other than those specified, with the
exception of transducers and cables sold by the manufacturer of the EQUIPMENT or SYS-
TEM as replacement parts for internal components, may result in increased EMISSIONS or
decreased IMMUNITY of the EQUIPMENT or SYSTEM.
d) The EQUIPMENT or SYSTEM should not be used adjacent to or stacked with other equip-
ment. IF adjacent or stacked use is necessary, the EQUIPMENT or SYSTEM should be
observed to verify normal operation in the configuration in which it will be used.
Guidance and manufacturer's declaration - electromagnetic emissions
The OM-4 is intended for use in the electromagnetic environment specified below.
The customer or the user of the OM-4 should assure that it is used in such an environment.
Emissions test Compliance Electromagnetic environment - guidance
CISPR 11 Group 1
The OM-4 uses RF energy only for its internal func-
tion. Therefore, its RF emissions are very low and
are not likely to cause any interference in nearby
CISPR 11 Class A
The OM-4 is suitable for use in all establishments
other than domestic and those directly connected to
the public low-voltage power supply network that
supplies buildings used for domestic purposes.
IEC61000-3-2 Class A