To maintain the safety and performance of the instrument, never attempt to do mainte-
nance of items other than those specified here unless done by an authorized service engi-
neer. For details about maintenance, read the chapter entitled "Maintenance" in this
REPLACING THE ILLUMINATION LAMP
The illumination bulb can be replaced if necessary. For specific instructions, see page 32.
REPLACING THE FUSES
Fuses on the primary side can be replaced if necessary. For specific instructions, see
Icon Prevention item Page
This instrument has been tested (with 120V) and found to comply
with IEC60601-1-2: 2001.
This instrument radiates radio frequency energy within standard
parameters and may affect other devices in the vicinity.
If you have discovered that turning on / off the instrument affects
other devices, we recommend you change its position, keep a
proper distance from other devices, or change the outlet.
Please consult the dealer from whom you purchased the instru-
ment for any additional questions. (for U.S.A. and Canada)
•TOPCON shall not take any responsibility for damage due to fire, earthquakes, actions
by a third party or other accidents, or the negligence and misuse of the user and use
under unusual conditions.
•TOPCON shall not take any responsibility for damage derived from the inability to use
this equipment, such as a loss of business profit and suspension of business.
•TOPCON shall not take any responsibility for damage caused by operations other than
those described in this Instruction Manual.
•Diagnoses made shall be the responsibility of the pertaining doctors and TOPCON
shall not take any responsibility for the results of such diagnoses.