16 SELECTING MODE
0-99 The time when EDM is cut off from distance
measurement is completed can be changed. This
function is effective for shortening the first time
measuring time when distance
measurement is started from distance
measurement completing state. (Default:3minutes)
0 :After completing distance measurement,
EDM is cut off immediately.
1-98 :EDM is cut off after 1~98 minutes.
99 :EDM is always switched ON.
0.2mm/1mm Select 1mm or 0.2mm for the minimum reading
unit in the distance mode (FINE mode).
FREE / HOLD Select Vertical angle setting in the Angle Offset
FREE: Vertical angle varies by the angle of the
HOLD: Vertical angle is fixed even if the angle of
the telescope changes.
Select the distance measurement mode when the
power is turned ON.
(Only for Laser
1 -99 Laser plummet function can be turned off
1-98 : Laser Plummet is cut off after 1 to 98
99 : Manual
ON / OFF Specify whether the buzzer sounds or not for every
horizontal angle 90°.
S/A BUZZER ON / OFF Specify whether the buzzer sounds or not in the set
OFF / 0.14 / 0.20 Set correction for refraction and earth curvature,
coefficient of refraction as ; K=0.14, K=0.20 or no
ON / OFF It is possible to retain the coordinate of instrument
point, the instrument height and prism height after
REC TYPE REC-A / REC-B Select REC-A or REC-B for data output.
REC-A : The measurement is made again and this
new data is output.
REC-B : The data being displayed is output.
CR,LF ON / OFF It is possible to output the data with carriage return
and line feed.
with RAW /
with RAW - 12Dig
Select to record coordinates in standard or with raw
ON/ OFF In the layout mode or data collect mode, it is
possible to record coordinates entered directly from
Select the displaying language.