6 SPECIAL MODE (Menu Mode)
6.1 Application Measurement (PROGRAMS)
6.1.1 Remote Elevation measurement (REM)
To obtain elevation of the point at which setting the target prism is not possible, place the prism at any
point on the vertical line from the target then carry out REM procedure as follows.
1) With prism height (h) input (Example :h=1.5m)
Operating procedure Operation Display
1After pressing the [MENU] key, press the [F4](P
key to get the menu on page 2. [MENU]
2Press the [F1] key. [F1]
3Press the [F1](REM) key. [F1]
4Press the [F1] key.
5Enter prism height. *1)
7Press the [F1](MEAS) key.
Measuring starts. [F1]
F3:Z COORD. P
R.HT = 0.000 m
--- --- [CLR] [ENT]
MEAS ––– NP/P –––
HD*[n] << m