6 SPECIAL MODE (Menu Mode)
6.1.2 Missing Line Measurement (MLM)
Measurement for horizontal distance (dHD), slope distance (dSD), elevation (dVD) and horizontal
bearing (HR) between two target prisms.
It is possible to enter the coordinate value directly or calculate from coordinate data file.
MLM mode has two modes.
1: MLM-1 (A-B, A-C) :Measurement is A-B, A-C, A-D,......
2: MLM-2 (A-B, B-C) :Measurement is A-B, B-C, C-D,......
zIt is necessary to set the direction angle of the instrument.
[Example] MLM-1 (A-B, A-C)
zProcedure of MLM-2 (A-B, B-C) mode is completely same as MLM-1 mode
Operating procedure Operation Display
1After pressing the [MENU] key, press the [F4](P
key to get the menu on page 2. [MENU]
2Press the [F1] key. [F1]
3Press the [F2](MLM) key. [F2]
4Press the [F1] or [F2] key to select using
[Example:F2 : DON’T USE]
F3:Z COORD. P