4 - 10
4.7 Selectting the Display
Five different displays can be selected. These are:
Channel number and measured values will appear on the display.
When an alarm occurs, type of alarm will appear.
Every channel will be shown for approximately 2 sec. If the input is greater than the
upper limit of the recording span, – – – – will appear.
If the input is less than the lower limit of the recording span, – – – – – will appear.
Measured Value (5 digits)
Type of Alarm (1 digit)*
Channel Number(1 digit)
*Type of alarm
: high limit
: low limit
: high difference limit
: low difference limit
DISPLAY MAN (except for the 1-pen model)
Channel number, measured values and type of alarm when it occurs will appear on the
display for a specific channel only. This display looks the same as the AUTO display.
The display is updated every two seconds for Pen models and at every measurement
period for Dot-printing model. Using the [CH UP] key, the displayed channel number
will go up.
Date, including the year will appear on the display. Leap years are provided.
Time will appear on the display.
Displays other than status displays will be turned off.
Selecting a screen
Pressing the [DISP] key cycles through the screens in the following order:
AUTO: automatic-mode screen
MAN: manual-mode screen
DATE: date-mode screen
TIME: time-mode screen
OFF: off-mode screen