CAUTIONS FOR USE
Before using, sterilize the sterilized cap, the front lens unit, the peripheral observation prism and the
anterior eye section observation lens.
Dispose of the instrument according to the local government's laws regarding disposal and recycling.
ENVIRONMENT FOR USE
Temperature : 10°C~40°C
Humidity : 30%~75% (without dew condensation)
Air pressure : 700~1060hPa
STORAGE, PERIOD OF USE AND OTHERS
1. Environmental conditions
Humidity: 30%~75% (without dew condensation)
Air pressure: 700~1060hPa
2. Storage requirements.
(1) Do not splash this instrument with water.
(2) Make sure this instrument is stored under safe conditions with regard to air pressure,
temperature, humidity, ventilation, sunlight or salty/sulfurous air.
(3) This instrument must be kept stable, without inclination, vibration or shock (not only in storage,
but also during transportation).
(4) Do not store this instrument in a location where chemicals are stored or gas is generated.
3. Limit of period of use
8 years following delivery, only if regular inspection and maintenance have been carried out
(according to self-certification [Topcon's data]).