PREPARATIONS BEFORE USE
SETTING ACCESSORIES: OFFISS, OFFISS Lite, Pro, Pro Lite
The illustrations display the OFFISS type.
1While pressing the electromagnetic lock release switch of the microscope operation handle,
move the 2nd arm to the upper limit position.
2Tighten and fix the 2nd arm lower limit knob by turning it clockwise at the lower limit lock position
WARNING Do not install/remove the accessories above the patient. An accessory
falling off could cause injury or even death.
WARNING Before installing the accessories, make sure all the arms are securely
locked. The 2nd arm may move suddenly, causing an injury.
WARNING After installing/removing the accessories, make sure that handles,
levers, knobs and rings are securely tightened. Any of these falling off
could cause injury or even death.
WARNING Before using this instrument, adjust the balance of the 2nd arm. The 2nd
arm may move up and down suddenly during an operation, causing an
WARNING Release the 2nd arm lower limit lock while holding it at the end. The 2nd
arm may move up and down suddenly, causing an injury.
CAUTION Do not open the instrument, as this may lead to an electric shock.
NOTE Ask your distributor or the Topcon offices stated on the back cover to repair the