PREPARATIONS BEFORE USE
6While pulling the 2nd arm lower limit lock release knob, raise the 2nd arm somewhat. The lock is
7Adjust the balance of the 2nd arm.
After installing or removing the accessories, the 2nd arm balance must be readjusted.
Refer to "ADJUSTING THE 2ND ARM BALANCE: Standard" on P.41.
INSTALLING THE INVERTER: OFFISS, OFFISS Lite
1Loosen the eyepiece lens/beam splitter fixing knob on the microscope unit and then remove the
eyepiece lens/beam splitter from the microscope.
2Fit the groove of the inverter to the pin on the top side of the microscope and tighten the eyepiece
lens/beam splitter fixing knob to fix the inverter.
We recommend using a hexagonal screwdriver.