PREPARATIONS BEFORE USE
INITIALIZATION: OFFISS, Pro
The illustrations display the OFFISS type.
1Press the initial switch on the front of the X-Y translator. The instrument is in the initial set condition.
Initialization cannot be carried out except the X-Y translator if the front lens is in use (or
after it has been removed). (Refer to "OPERATION OF THE FRONT LENS: OFFISS,
OFFISS Lite" on P.70.)
Initialization can also be carried out with the foot switch. If you wish to use this foot switch
operation, you must change the setting. Please ask your distributor to carry out the change.
CAUTION Before moving the instrument, make sure there is no-one and nothing
within collision range. Injury may result from the instrument colliding with
X-Y translator Center position
Focusing unit Approx. 35mm from the upper limit
Approx. 25mm from the lower limit
ZOOM Minimum magnification
Rough focusing unit Approx. 40mm from the upper limit
Approx. 20mm from the lower limit