PREPARATIONS BEFORE USE
INSTALLING THE STERILIZED CAP: Standard
Sterilize the sterilized caps and then place them in the specified positions. The following table displays
the setting positions of the sterilized caps.
* and are attached to only the type equipped with the
changeable beam splitter.
ADJUSTING THE PUPILLARY DISTANCE: Common
1Adjust the pupillary distance by looking through the eyepiece lenses and turning the pupillary
distance adjustment knob until both eyes are able to see correctly.
Record the read value. Then set the pupillary distance at the value.
ADJUSTING THE OBSERVATION ANGLE: Common
1Hold the observation angle handle and adjust it to the required observation angle.
Type Position Quantity
Optical unit inclination knob 1
Pupillary distance adjustment knob 2
Spectral ratio knob of changeable beam
BMicroscope operation handle 2
Observation angle handle 1
Spectral ratio lever of changeable beam
Type with the changeable beam splitter