OPERATIONS IN USE
PROCEDURE IN EMERGENCY: Pro, Pro Lite
In an emergency, follow the check list below and contact your distributor. If you do not encounter your
problem in the "Status" column, please contact your distributor.
The inverter does not work. The manual IN/OUT knob (with sterilized cap) is located on the
front of the inverter. Turn this knob clockwise to the “click”
Zoom mechanism does not work. Adjust the zoom manual knob with a coin, etc.
An error is displayed. Press the EXIT switch on the 1st arm panel to reset.
The focusing unit does not work. Move the 2nd arm up and down for focusing.
Electromagnetic lock is not released. Push each arm firmly to move it.
The illumination angle switch does
Adjust the illumination angle manual knob with a coin, etc.