AFTER USE / MOVING AND STORING: OFFISS OFFISS Lite, Pro, Pro Lite
AFTER USE: Common
1Turn off the power switch.
2Remove the power plug from the power outlet.
3Carry out cleaning.
Refer to "CLEANING THE OBJECTIVE LENS / THE EYEPIECE LENS" on P.101.
Refer to “CLEANING THE FRONT LENS STORING UNIT” on P.101.
Refer to "CLEANING OTHERS EXCEPT OPTICAL SYSTEM" on P.102.
MOVING AND STORING: OFFISS, OFFISS Lite, Pro,
WARNING Make sure no-one is too close to the instrument before moving the arm.
Anyone touching the instrument may be injured.
WARNING Before moving the instrument, make sure that no-one is near. Then,
move it carefully. Injury could be caused by the instrument touching
WARNING Hold the microscope operation handle while pressing the
electromagnetic lock release switch to unlock it. The arm may rotate or
move up and down suddenly, causing an injury.
CAUTION Take care when moving this instrument through a door or in a room with
a low ceiling. If the top of this instrument collides with something, it could
CAUTION Watch out for devices, beds, walls, etc. in the room. If the instrument
collides with anything, serious breakage could result.
CAUTION Watch out for stairs and uneven floors. This instrument may tip over.
CAUTION Watch out for slopes. Due to its increased speed, the instrument may
get away from you on a slope.
CAUTION Do not install this instrument on an incline. This may cause it to move