MOVING AND STORING: OFFISS, OFFISS Lite, Pro, Pro Lite / MOVING AND STORING: Standard
9Hold the base movement handle and move the instrument gently.
10 Lower the caster lock lever to lock it when reaching the destination or storing place.
MOVING AND STORING: Standard
Please carry out the following procedures when moving and storing this instrument.
1Loosen the 2nd arm vertical movement fixing handle, the 2nd arm rotation fixing handle and the
1st arm rotation fixing handle. Move the 1st and 2nd arms so as not to hinder storing and then
tighten and fix them.
The 2nd arm can be fixed at the lower limit position by setting the 2nd arm lower limit lock.
(Refer to "LOCKING/UNLOCKING THE 2ND ARM LOWER LIMIT: Standard" on P.50.)
WARNING Before moving the instrument, make sure that no-one is near. Then,
move it carefully. Injury could be caused by the instrument touching
Always hold the microscope operation handle when loosening either the
2nd arm vertical movement fixing handle, the 2nd arm rotation fixing
handle or the 1st arm rotation fixing handle. The arm may rotate or move
up and down suddenly, causing an injury.
CAUTION Take care when moving this instrument through a door or in a room with
a low ceiling. If the top of this instrument collides with something, it could
CAUTION Watch out for devices, beds, walls, etc. in the room. If the instrument
collides with anything, serious breakage could result.
CAUTION Watch out for stairs and uneven floors. This instrument may tip over.
CAUTION Watch out for slopes. Due to its increased speed, the instrument may
get away from you on a slope.
CAUTION Do not install this instrument on an incline. This may cause it to move