BASIC SPECIFICATIONS: Common
* Subject to change in specifications and appearance for future improvement without advance notice.
This instrument complies with IEC60601-1-2: 2001.
a) MEDICAL ELECTRICAL EQUIPMENT needs special precautions regarding EMC and needs to be
installed and put into service according to the EMC information provided in the ACCOMPANYING
b) Portable and mobile RF communications equipment can affect MEDICAL ELECTRICAL EQUIPMENT.
c) The use of ACCESSORIES, transducers and cables other than those specified, with the exception of
transducers and cables sold by the manufacturer of the EQUIPMENT or SYSTEM as replacement
parts for internal components, may result in increased EMISSIONS or decreased IMMUNITY of the
EQUIPMENT or SYSTEM.
d) The EQUIPMENT or SYSTEM should not be used adjacent to or stacked with other equipment. If
adjacent or stacked use is necessary, the EQUIPMENT or SYSTEM should be observed to verify
normal operation in the configuration in which it will be used.
e) The use of the ACCESSORY, transducer or cable with EQUIPMENT and SYSTEMS other than those
specified may result in increased EMISSION or decreased IMMUNITY of the EQUIPMENT or
Inverter 4582610000 MS-IN01-BU
Cable 4582615100 INV CABLE
Microscope type Galileo type
Magnification change type Electric zoom continuous change
Eyepiece (Eyepiece magnification) 12.5×
Objective lens f = 200mm
Display magnification (×) 4.2/5/6/7/8/9/10/11/13/15/17/19/21
Total magnification (×) 4.2 ~ 21×
1st arm length (Distance between shafts) 375mm
1st arm rotation range 300°
2nd arm length (Distance between shafts) 990mm : OFFISS, Pro
875mm : OFFISS Lite, Pro Lite, Standard
2nd arm rotation range 300°
2nd arm vertical movement range 600mm
2nd arm mounting weight 6 ~18kg : OFFISS, Pro
9 ~21kg : OFFISS Lite, Pro Lite, Standard