ArtE Approach Rate Time: Select OFF, 0.01 to 99.99 minutes. The function
defines the amount of Rate applied when the input is outside of the
Proportional Band. The ArtE time and the rtE time are independent and have
no effect on each other. To increase damping effect and reduce overshoot
set the approach rate time for a value greater than the natural rise time of
the process (natural rise time = process value time to set point).
Fint Fuzzy Logic Intensity: Select 0to 100% . 0% is OFF (disables Fuzzy Logic).
The function defines the amount of impact Fuzzy Logic will have on the out
put. If Fint is set to 0, Fbnd and Frte below will not appear.
Fbnd Fuzzy Logic Error Band: Select 0to 4000 °F, °C, or counts. Sets the band
width of the Fuzzy Logic. Set Fbnd equal to PID proportional band (Pb1)
for best results. Will not appear if Fint is OFF.
FrtE Fuzzy Logic Rate of Change: Select 0.00 to 99.99 counts/second. For best
initial setting, find the count/second change of process value near set point
1 with output ON 100%. Multiply this value by 3. Set FrtE to this
calculated value. Will not appear if Fint is OFF.
PEA The Peak feature stores the highest input the control has measured since
the last reset or Power On. At Power On PEA is reset to the present input.
To manually reset the value PEA must be in the lower display. Press the
ENTER key to reset. PEA will be reset and display the present input value.
UAL The Valley feature stores the lowest input the Instrument has measured
since the last reset or Power On. At Power On UAL is reset to the present
input. To manually reset the value UAL must be in the lower display. Press
the ENTER key. UAL will be reset and display the present input value.
PctO Percent Output Feature: Select On or OFF.
On When selected On, the HOME lower display will indicate the output
of the controller in percent. An “o” will appear in the right hand side
of the lower display to indicate percent output for SP1. An "o" will
appear on the right hand corner of the lower display to represent
percent output for SP2, if the control is so equipped.The display will
alternate between these values.
OFF Percent Output display is disabled.
Prog Ramp/Soak Feature: Select On or OFF
On Allows Programmed Ramp/Soak function to be started by the
Run/Hold key on the control front panel.
OFF Turns Ramp/Soak function OFF and resets program to beginning.
PSEt Programmer function set. Select On or OFF.
OFF Skip Ramp/Soak Programming. Go to next Secondary Menu Item,
In PC on the next page.
On Enable Ramp/Soak Programming.