Refer to Figure 2-1 for outline dimensions. Drill appropriate
mounting holes as required.
Refer to Figure 2-2, wiring installation drawing for appropriate
terminals for interconnections. The signal leads from the turbine
should be shielded whenever the FLSC-18B is not integrally
mounted. Ground shield on one end only. If desired,use solder
lug provided on ground stud. Connections to the terminal block
should be carefully dressed to avoid having bare wires
extending pass the screw clamp on the terminal block. This is
particularly important or units mounted within the explosion
proof enclosure. Wires should be neatly dressed near bottom of
enclosure to assure wiring will not become fouled when cover is
Connect two conductor shielded cable from flowmeter. Connect
shield to FLSC-18B only.
Unit is powered by a DC voltage which must be within 8 to 40
volts. Connect power as outlined in Figure 2-2.