2.2 INSTALLATION OF THE FLSC-51/51B
The FLSC-51/51 B should be placed in a convenient location
which maintains access to the unit should repairs or
readjustment be required.
If the unit is mounted on a turbine flowmeter in an explosion
proof conduit enclosure, orientate the flowmeter and run
conduit to prevent the accumulation of moisture in the conduit
enclosure as much as is practical. In addition, assure electronics
will not be overheated by hot process lines.
Refer to Figure 2-1 for outline dimensions. Drill appropriate
mounting holes as required.
Refer to Figure 2-2, wiring installation, for appropriate terminals
for interconnections. The signal leads from the turbine should be
shielded whenever the FLSC-51/51B is not integrally mounted.
Ground shield on one end only If desired,use solder lug
provided on ground stud. Connections to the terminal block
should be carefully dressed to avoid having bare wires
extending pass the screw clamp on the terminal block. This is
particularly important for units mounted within the explosion
proof enclosure. Wires should be neatly dressed near bottom of
enclosure to assure wiring will not become fouled when cover is
installed. Connect two conductor shielded cable f rom
flowmeter. Connect shield to FLSC-51/51B only.
Unit is powered by a DC voltage, 8-35 Vdc. Connect power as
illustrated in Figure 2-2.
Figure 2-1. Outline and