HOW TO INSTALL THE INSTRUMENT
1 Fasten the clamping knob.
2 Hold the instrument body firmly at the specified positions and place it on the auto-
matic instrument table.
For the automatic instrument table, see “OPTIONAL ACCESSORIES” on page 46.
3 After installing the instrument, loosen the fixing knob.
Now the body components can be moved.
4 If the machine body is slightly off level, properly turn the adjusters at the four corners
for fine adjustment.
Do not unscrew the adjusters more than 1cm.
Before carrying the instrument, be sure to affix it firmly by
turning the fixing screw at the base.
If the instrument is moved with the screw loosened, it may
result in damage to the instrument.
When moving the instrument, be sure to hold it at the bottom sur-
face with two people. Carrying by one person may cause back
injury or injury by falling parts. Also, holding areas other than the
bottom surface may cause pinching fingers between parts and
injury by falling parts as well as damage to the instrument.
To avoid potential injury, hold the instrument in the proper
Holding the instrument Specified holding positions