During the initial setting, date, time, operating time of the power save function, RS-
232C, mode of average value, buzzer and message can be set.
1 Make sure the power cable is connected.
For connection, see “HOW TO CONNECT THE POWER CABLE” on page 14.
2 Check the no-patient condition of the instrument and turn the POWER SWITCH ON.
Displaying The Menu Screen
1 Make sure that the Measurement screen is displayed.
2 Press on the control panel.
The Menu screen is displayed.
Returning To The Measurement Screen
1 Press , on the control panel, move the cursor to “EXIT” and press
MEASUREMENT SWITCH .
Or, press on the control panel. The Measurement screen returns.
• When the machine is moved from a cold room to a warm room or
when the room temperature suddenly rises, it may cause dewing in-
side the machine and disable measurement. In this case, leave the
machine alone for about 30min until it reaches room temperature.