3 Hold the paper and pull it diagonally to cut.
HOW TO CORRECT MEASUREMENT VALUES
Though the machine is adjusted for displaying optimal measurement values, the values
can be corrected within a -4 ∼ +3mmHg range.
1 Make sure the power is OFF.
2 Open the control panel lid.
3 Using a screwdriver, turn the “ ” of the select switch.
Settings: 0: Base setting For shipment, the switch is set here.
1: +1mmHg Base setting + 1mmHg
2: +2mmHg Base setting + 2mmHg
3: +3mmHg Base setting + 3mmHg
F: -1mmHg Base setting - 1mmHg
E: -2mmHg Base setting - 2mmHg
D: -3mmHg Base setting - 3mmHg
C: -4mmHg Base setting - 4mmHg
4 Turn the POWER SWITCH ON.
The measurement procedure is exactly the same as with the correction.
• To avoid paper jams, cut the paper carefully and evenly.
Never set the select switch for more than eight points to avoid