This product conforms to the EMC Standard(IEC 60601-1-2:2001).
a) MEDICAL ELECTRICAL EQUIPMENT needs special precautions regarding EMC and needs to
be installed and put into service according to the EMC information provided in the ACCOMPA-
b) Portable and mobile RF communications equipment can affect MEDICAL ELECTRICAL
c) The use of ACCESSORIES, transducers and cables other than those specified, with the excep-
tion of transducers and cables sold by the manufacturer of the EQUIPMENT or SYSTEM as re-
placement parts for internal components, may result in increased EMISSIONS or decreased
IMMUNITY of the EQUIPMENT or SYSTEM.
d) The EQUIPMENT or SYSTEM should not be used adjacent to or stacked with other equipment.
IF adjacent or stacked use is necessary, the EQUIPMENT or SYSTEM should be observed to
verify normal operation in the configuration in which it will be used.
e) The use of the ACCESSORY, transducer or cable with EQUIPMENT and SYSTEMS other than
those specified may result in increased EMISSION or decreased IMMUNITY of the EQUIP-
MENT or SYSTEM.
Item Article code Model No. Length(m)
RS-232C CROSSING CABLE
- - 3.5
Guidance and manufacturer's declaration - electromagnetic emissions
The CT-80 is intended for use in the electromagnetic environment specified below.
The customer or the user of the CT-80 should assure that it is used in such an environment.
Emissions test Compliance Electromagnetic environment - guidance
Group 1 The CT-80 uses RF energy only for its internal
function. Therefore, its RF emissions are very low
and are not likely to cause any interference in
nearby electronic equipment.
Class A The CT-80 is suitable for use in all establishments
other than domestic and those directly connected
Class A to the public low-voltage power supply network that
supplies buildings used for domestic purposes.