Items of setting (5 items)
⋅ Machine No. (EQUIPMENT) Set value: 0∼99. (For shipment, “1” is set.)
When more then one unit of inspection equipment
is installed in the same hospital, for example, data
can be controlled using these Machine Nos.
⋅ Input ID mode (ID MODE) Set value: 1 or 2 (For shipment, “1” is set.)
1. For inputting patients' ID numbers.
2. For inputting the temporary ID numbers of new
⋅ Work ID No. Set value: 0∼9999 (For shipment, “0” is set.)
(WORK ID NO.) Desired serial numbers can be attached to measure-
ment results. The number is automatically added (1
at a time) each time the printing of data communica-
tion is performed.
⋅ Communication format Format: OFF, MODE 1, MODE 2, MODE 3, MODE 4,
(FORMAT) MODE 5, STD 1, STD 2 (For shipment, “OFF” is set.)
MODE 1: When is pressed, communication is done
MODE 2: Data communication is done every measure-
MODE 3: When is pressed, communication is done
MODE 4, MODE 5: When is pressed, communica-
tion is done without printing.
STD 1, STD 2: When is pressed, communica-
tion is done after printing.
⋅ Communication speed (SPEED) Baud rate: 2400, 9600 (bps) (For shipment,
2400bps is set.)
Under the RTS-CTS control, if no CS (CTS) signal is returned from the receiver side,
transmission can be canceled by pressing . Also, if no DR (DSR) signal is re-
turned, it is recognized as a communication failure and FAIL is displayed on the moni-
* Certain models may not export data of MODE5 & STD2. It indicates settings as
When Mode 1, 2 or 3 is set, data is transmitted automatically without confirmation
from the receiver side. When Mode 4, Mode 5, STD 1, STD 2 is set, communication is
controlled by RTS-CTS.