1.3.6 HOW TO TURN POWER ON/OFF
Push the power switch to the ON position to turn on power.
The mark [CAL.] is displayed and the zero-adjustment starts.
After zero-adjustment complete, the current measured value appears.
・ Response switch can be selectable before and after zero-adjustment.
Response switch ☞ ‘1.2 Names and functions of parts’
・ The time required to complete zero adjustment differs depending on the setting
of the response switch. FAST: about 15 seconds; SLOW: about 50 seconds.
・ No Cap is required in Zero adjustment.
To turn off power, push the power switch to the OFF position.