・ Enter the correction factor in the range 0.001 to 1000.
・ To cancel the correction factor, enter 1.000 as the correction factor.
・ After completion of the C.C.F. setting mode, the system returns to the previous
・ Once correction factor is activated, Readout is multiplied by the correction
Readout = raw measured value x Correction factor (C.C.F.).
・ If the readout exceeds 999900 due to entering too large a correction factor, the
code E4 appears.
・ During measurement of the integral illuminance, the correction factor cannot be
・ Correction factor data is saved after turning off.
3.4.1 SETTING PROCEDURE (Ex:To enter 1.2)
The currently entered correction factor appears for 3 seconds after pushing [C.C.F.]
Push the [C.C.F.] switch again within 3 second after the above process, then the
screen turns to the screen for numeric value entry.
Push the [0 - 9] switch to change the blinking digit to 1.
Push the [INV.] switch, and check if the mark INV appears.