7. APPLIED OPERATION AND SETTING OF VARIOUS FUNCTIONS
1Set up the RL-SV1S on Point A of the base line using the plumb bob on the tripod.
2Using the sight at the upper section of the RL-SV1S, adjust the direction on top of the tripod
and roughly align the Y+ direction to Point B on the standard axis.
3Horizontally rotate the laser beam of the RL-SV1S. (X+0.000%)
4At Point B adjust the height of the LS-80L installed on a pole, align the standard position of
the LS-80L with the laser beam and fix.
5Set the RL-SV1S at X+3.000% grade.
6Align the RL-SV1S direction on top of the tripod so as to have the laser beam in the on-
grade position of the LS-80L in step 4.
• Do not change the height of the LS-80L installed on the pole.
If the height of the RL-SV1S is changed, return to step 3 and redo the adjustment.