7. APPLIED OPERATION AND SETTING OF VARIOUS FUNCTIONS
[Display during the Matching Mode]
is flashing: in the Matching Mode. It is possible to adjust the grade of the laser beam
with arrow keys. When a certain amount of time has passed after the arrow key operation, the
light that was flashing will remain lit.
Grading cannot be adjusted when is on.
: shows the axis on which the grading is being adjusted.
SLOPE: shows the axis that is being graded. Auto leveling of the axis will not function at this
LEVEL: shows the axis whose grade is being automatically leveled horizontally. A flashing
display indicates that auto leveling is being performed.
The height alert function setting is valid at this time.
To readjust grading, return to step 5 and follow directions thereafter. However, skip steps 7 and
8 for an axis that is already graded.
Exiting the Matching Mode
When is flashing: press the key to exit the Matching Mode. Press key to set
the grade value.
When is lighted: press the key to set the grade value.
How to set grade (p. 32)