7. APPLIED OPERATION AND SETTING OF VARIOUS FUNCTIONS
The laser beam can be moved to the direction of the key during vertical rotation.
• Only the X axis can be moved.
• Allowable alignment range: ±5° (when the instrument is set up on the 0° surface)
1Set the instrument.
2Press the Power control key to turn unit on. When
auto leveling is complete, the laser beam will be
3Move and set the RL-SV1S to align reference point
A and the laser beam.
• Make sure that the RL-SV1S is set so that the
bubble is at the center of the vertical rotation
circular level vial on the control panel.
7.2 Line Control (manual vertical beam alignment)
If it is unable to visually confirm the laser
beam on reference point A and B, set up the
LS-80L on either of the Points.