7. APPLIED OPERATION AND SETTING OF VARIOUS FUNCTIONS
Masking (Laser beam shutter) setting
Depending on the status of the location where the instruments are used, laser beam emission
to unnecessary direction can be shut off.
1Press key to display the menu screen.
2Use the key to position on the Mask display and press the key.
3Select the direction you desire to mask using the arrow keys.
Each press repeats mask activating/releasing.
surface diagram and
Arrow keys and masking
The state when masking is not
activated. (Laser beams are emitted
to all directions.)
Displays the masking direction
The status in which the Y+ direction
(Laser beam is shut off in the Y+